'Homemade Microscope'  by  Roger C. Baker
This compelling article  [  7.2 MB,  12 pages]  speculates that Leeuwenhoek must have invented a compound microscope.
Roger describes how he thinks it was done, along with construction details for a hobbyist 'compound lens' microscope.
With permission from Science Probe April 1991
LED illuminator for reflected light microscopy and photomicroscopy
Roger C. Baker,  March 10, 2005.
This is a view of the "top" of my illuminator.
[Click on photo for fullsize view]
This photo shows the illuminator mounted on the 4x objective of my Chinese biological microscope.
The device was made to fit on my 4x and 10 x planar microscope objectives, which are primarily designed to view slides. Accordingly, they have a relatively short focus and high numerical aperture. When the 4x objective is focused on an object, there is about 8 mm of clearance space. The LED illuminator resembles a removable cylindrical brass collar that clamps around the microscope objective. This collar holds four or more wooden fin-like LED supports. The LEDs are each wired in parallel with current limiting resistors via two semicircular copper wire connectors.
The collar was made from a strip of brass sold for
hobbyists (K&S Engineering, Chicago) 0.025” thick and 1" wide, and about 4.5” long.
The strip width should be trimmed if necessary to slightly less than the flat cylindrical length of the objective. The length of the brass strip should be such that it equals the circumference of the objective lens plus the length of two extended 5/8” flanges used to tighten it.
The brass strip should be initially heated to a red heat with a propane torch to anneal and soften it. The two ends are then bent sharply upward in the same direction at a 90 degree angle. Finally, the strip is bent in the opposite direction around a metal rod, with about the same diameter as the objective. I tapped the strip with a soft piece of wood so it conforms closely to the rod to make a nearly closed cylinder leaving the flanges extending outward and parallel to each other. The resulting shape should be such that there is some clearance around the objective until the extending flanges are squeezed together, causing the springy collar to tighten and grip the objective. Finally the flanges are clamped together and a dimple is made near to their edge with a nail set. A hole is drilled through the both of them them, allowing a nut and bolt to be inserted. These are used to tighten the collar onto the objective using mild finger pressure.
The white LEDs I used have a beam spread of about 15 degrees, and are said to generate about 10 candellas apiece (RL5-W10015). Not including shipping, they cost $1.19 apiece from:   http://www.superbrightleds.com/leds.htm
Roger Baker - Chalky Limestone Slab
Here is an image of my illuminator as used to show a chalky limestone slab stained with gentian violet
(sometimes called crystal violet). The view is a ground and stained fossil
surface as seen at 40 power magnification.|
Note: Click on image for fullsize view.
There are a few artifacts; the light
slash to one side is the reflective meniscus of the cover glass edge, there are
a few bits of stray dust on the glass, etc.|
The rock sample is typical of Austin area sedimentary limestone, with a flower-like cross section of a fossil visible toward the bottom center of the image.
My Canon Powershot A-80 4 megapixel digital camera was held in direct contact with the rim of the wide glass binocular eyepiece and adjusted sideways until the image was nice and bright on the camera's LCD display.
A Previously Unpublished Limestone Micro-Fossil Staining Procedure
The surface of the rock is roughed out with coarser grinding methods and finally ground against glass with
# 600 silicon carbide/water slurry. It is rinsed clean with fresh water. A few
drops of a strong alcoholic solution of gentian violet dye is then spread across
the surface with the help of the shaft of the eye dropper. It is then allowed to
dry, giving the rock surface an iridescent green luster. |
stain is next rinsed off with several copious rinses of fresh water, without
mechanical wiping although a hand-spray bottle can be used, to remove most of
the non-absorbed stain. A drop of glycerine is then applied to the wet stained
surface. Finally a cover glass placed on top of the glycerine to give an
optically flat surface to view or photograph.
The nature of the gentian violet staining process at near neutral PH seems to depend on the surface chemistry of the exposed material, and not merely the porosity of the mineral. That the atomic surface layer of an insoluble mineral can be stained is indicated to me by the fact that the glassy fractured grains of silicon carbide abrasive, which is a hard and chemically inert crystal, become stained and then look somewhat like flakes of purple confetti.